EXFO OTDR 2 • User guide • Download PDF for free and without registration!. Get EXFO OTDR 2 Owner’s manual. Get all EXFO manuals!. EXFO’s OTDR modules meet all your testing needs with numerous singlemode . Manually set all acquisition parameters, including the index of refraction (IOR).
|Published (Last):||25 January 2004|
|PDF File Size:||10.56 Mb|
|ePub File Size:||9.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
OTDR and iOLM
Push the connector otvr so that the fiber-optic cable is firmly in place, thus ensuring adequate contact. From the Main Menu, select the Reference button. Keep the unit out of direct sunlight. Customizing Your OTDR Selecting the Default View Selecting the Default View You can select which view will be displayed by default once all the acquisitions are performed at all the selected wavelengths and mamual analysis of the last wavelength is complete.
One field-optimized, powerful test solution. Page 84 Customizing Your OTDR Selecting the Default View Selecting the Default View You can select which view will be displayed by default once all the acquisitions are performed at all the selected wavelengths and the analysis of the last wavelength is complete.
Page 63 Testing Fibers Defining Launch and Receive Fiber Settings The application allows you to manually set the lengths or the events of both your launch and receive fibers.
If you want to create a folder, you will have to do it manually via File Manager available from the Home screen. Three traces are shown, corresponding to three threshold level settings.
To decrement values, the start number must be higher than the stop number. To revert to the complete graph view: One ” ” is the default format. When the application prompts edfo, select Yes.
Pdf Download | EXFO OTDR 2 User Manual ( pages)
Move marker A to the beginning of the event. If you want to modify the parameters, enter the desired values for the current trace in the appropriate boxes. The signal strength is measured for specific intervals of time and is used to characterize events. From the Wavelengths list, select the pair of wavelengths for which you want to define the delta value. Tap Insert to insert the event or Cancel to return to the measure tab without making any changes.
Tell us about it. Carefully align the connector and port to prevent the fiber end from touching the outside of the port or rubbing against other surfaces.
The area between markers A and a must not include any significant variation. Creating and Generating Reports Generating a Report 3.
Support personnel will determine if the equipment requires service, repair, or calibration.
EXFO OTDR 2 User Manual
From the Main Menu, select Test Configuration. Tap OK to confirm you selection. Select the boxes corresponding to the desired test wavelengths. By default, all available test wavelengths are selected. The attenuation, 4 point event loss, and maximum reflectance are also displayed. Detected events are distinct even if they cover more than one point on the trace.
However, you can also set them at a later time ,anual in that case, you have to reanalyze the trace if you change the backscatter coefficient see Viewing and Modifying Current Measurement Settings on page EXFO shall not be liable for damages resulting from improper usage or unauthorized modification of the product, its accompanying accessories and software.
The trace display and events table are explained in the following sections. Hiding the fiber sections will not delete these items. Under Detection Thresholds, enter the desired values in the appropriate boxes.
EXFO OTDR 2 User Manual | pages
For more information on positioning markers, see Using Markers on page The table below presents the possible behavior of the zoom and markers when you open traces. Under Advanced Parameters, to enable the first connector check, select the First connector check box. From the Main Menu, tap Identification. If former results have not been saved yet, the application prompts you to save them before starting a new exfl. The available choices are: